The TE 58 is a hermetically sealed high-pressure rotary tribometer, designed to operate at pressures up to 15 MPa with conventional refrigerants or CO2 as the pressurizing gas/fluid.
The TE 58 has servo controlled loading with force transducer feedback, servo motor with encoder feedback for speed control, SUPERSLIM Serial Link Interface Module and COMPEND 2000 Window compatible control and data acquisition software. The machine is floor-standing.
Motor and Test Spindle
The test spindle, which projects downwards, is fully enclosed within the pressure vessel and incorporates hybrid bearings that run in the working atmosphere. The spindle is connected to the drive motor via a magnetic drive coupling. Moving specimen tooling is attached to the lower end of the spindle.
Load Application and Measurement
Load is applied by means of a servo controlled ball screw actuator, mounted beneath the pressure vessel. A rod extension projects upwards into the vessel and is sealed by means of a high pressure stainless steel bellows. The ball screw actuator thus has to have the capacity not only to apply the required load within the vessel, but also to overcome the pressure forces acting on the bellows.
Fixed Specimen Assemblies and Friction Torque Measurement
The fixed test assemblies are torque reaction mounted on top of the spring-pack assembly. Friction torque is measured by means of a torque arm connected to a pressure compensated force transducer.
Access to Test Assemblies
In order to gain access to the test assemblies, the chamber is split at its mid-point and the upper half of the chamber then raised on pneumatic bellows. Standard ports are provided for charging and scavenging the chamber.
Control and Data Acquisition
The TE 58 has PC based sequence programmable control and data acquisition. This is provided by an integrated Serial Link Interface Module and COMPEND 2000 software running on a host PC, operating under Windows. Data is stored to hard disc in standard spread sheet compatible file formats (.csv or .tsv).
Tests are defined by a sequence of steps, each step containing set-point, data recording rates and alarm level information. Set-points may be adjusted by step change or ramp. The test sequence is followed unless interrupted by the operator or an alarm. Set-points may also be adjusted manually using on screen toggles.
Sliding Four Ball Assembly
Three Pin /Ball on Disc
Vane Pump Assembly
Thrust Washer Assembly
Maximum Torque: 10 Nm Maximum Load: 5000 N Maximum Pressure: 15 MPa Maximum Speed: 2000 rpm Maximum Temperature: 120 C Environment: CO2, R407, R410 Motion: Continuous sliding Interface: Serial Link Interface Module Software: COMPEND 2000 Controlled Parameters Rotational Speed Temperature Load Test Duration Recorded Parameters Rotational Speed Friction Torque Temperatures Number of Revolutions Test Duration Sliding Speed Friction Coefficient Sliding Distance Services Electricity: 220/240V, single phase, 50 Hz, 5.5 kW 110/120 V, single phase, 60 Hz, 5.5 kW Clean, dry air: 4 cfm at 8 bar (120 psi)
ball on disc compressor materials high pressure testing refrigerant lubricants sliding four ball vane pump
Download the Machine Leaflet